Got my Nano v2 today… Amazed how small, yet readable it is… I put it through its setps and
upgraded to v3.43 and everything is working great…
I do have a question though, I am upgrading to x10 probes (via BNC converter) and I got the following
x1 display (1v/div 100us/div)
Looks fine, is there a way to get rid of the little peaks? calibration?
x10 display (2v/div 200us/div)
Now, this by no way looks right, when I adjust the screw on the probe, it will just flatten
out the top and bottom. Do I need to perform any adjustments on the Nano besides
changing the attn to x10?