i have a dso nano V1 that doesnt seem to be working. i cannot get it to read anything.
however, the signal generator output works. this has been confirmed by using another scope to measure it.
also confirmed that the probe is good.
could the firmware need to be reloaded? (i believe it is ver 2.2, but i dont have the scope infront of me so im not 100% sure), or is the scope most likely broken?