Took V2.4 to my work (NIST traceable cal lab) today.
Horiz timing was perfect, literally less than 1%.
Verticle accuracy was about 2% on all ranges.
Bad news is vertical bandwidth was not even close to 1Mhz. Mine barely makes 400Khz.
Will re-try monday to see if it’s the crappy probe or what.
Low sample rate severely chopped the sine wave above 400Khz.
I especially like the FIT mode as it autoranges vertically and horizontally like an autorange DVM.