ok, i have had a few days to mess around with this device… And it is a good little scope.
I have a quesiton on calibrating the v2, I know the basics of having the trace centered, etc…
But, with that being said, if I go to “ME” and look at the data, I am getting variable
numbers in the vpp / Vmin, Vmax & Vavg. How does one go about getting those to ZERO?
Would that not be accurate?
It also seems that when I messed around with the cal off and cal gain a little (not having a clue what I was doing)
The trace line that HAD a little noise was completely flat. ?
Also, once I shut the unit off and turn it back on, I have to move the V pos back to center
Is there a way to save my setup?
Just looking to make it more accurate…